Last Updated on March 23, 2023 by You Ling
Frequently Asked Questions 1: Ultrasonic vibrators are damp, which can be judged by inspection of insulation resistance. If the resistance is not more than 5 megohms, then the transducer is wet, and it should be dried. The same is true for ultrasonic logging transducers.
Frequently Asked Questions 2: Ultrasonic transducers, or ultrasonic logging transducers, have cracked or broken ceramic materials. In this case, if it is individual, disconnect the damaged vibrator to avoid affecting the normal vibrator. usage of. If it doesn’t work, it’s an overall replacement. In addition, if the vibrator is degummed, it needs to be processed in time to avoid affecting the normal use of the transducer.
For ultrasonic transducers, as well as ultrasonic logging transducers, the two aspects mentioned above are very important and fundamental, and also for us, especially learners, must be carefully studied and firmly mastered. In practice, it is good to apply this kind of transducer to get good results.